ION7650 METER PDF
◇During normal operation of the ION / ION meter, hazardous voltages are present on its ter- minal strips, and throughout the connected potential. PowerLogic ION Used at key distribution points and sensitive loads, ION and ION meters offer unmatched functionality including advanced . Download: PowerLogic™ ION / ION Meter User Guide – PowerLogic ION/ION,PowerLogic ION RTU.
|Published (Last):||15 May 2008|
|PDF File Size:||19.97 Mb|
|ePub File Size:||17.92 Mb|
|Price:||Free* [*Free Regsitration Required]|
Modem RI Green Indicates a ring is detected by the modem. Auto, 10BT hal f, 10BT full.
Location Front of met er. Refer to Step 8 to learn how to configure Volts Mode on the meter. View Meter Data Use the softkeys to select which data to display. Pin 3 – T ransmit Data – Pin 2.
Maximum Cable L ength m ft total for entire bus. Name Plt Nameplate information.
PowerLogic™ ION / ION Meter User Guide | Schneider Electric
Open the PDF directly: For typical installations, the battery should last 20 years or more. PT and CT secondary circuits ar e capable of genera ting.
Timing Resolution 1 ms. Maximum Devices per bus The light should always remain on when the meter is in service. MasterDNP V3. T ransformer on Vc Normal or Inverted Normal. Applying current levels incom- patible with the current inputs will permanently damage the meter. External V oltage VDC continuous. To select another language:. Wire T ype The LEDs on the front panel indicate iom7650 he following:. These limits are designed to provide reasonable protection against harmful interference when the equipment is operated in a commercial environment.
Lat e nc y min. Wye Star wiring for PT primaries and secondaries. Connector T ype RJ Use PTs for higher volt ages. Power up the meter. Null modem cable pinout. Wire T ype High quality Category 5 or 6 unshie lded twisted pair cable.
Operation of this eq uipment. Top green LED indicates the meter is operational. T ransformer on I5 Normal or Inverted Normal. D Meteg Digital Inputs. Internal Excitation Exter nal Excitation. Basic Ion760 — Rear View.
Wire the Power Supply Step 7: Fac tor y ION. English, Spanish, F rench. Slide the four brackets into their slots on the back of the meter and secure using the four Phillips head screws. Do not power up the met er until the curre nt and voltage wiring is completed. Form A Digital Outputs: Starting Curre nt 0.