BRAND TRANSFERPETTE MANUAL PDF
BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||16 April 2007|
|PDF File Size:||19.42 Mb|
|ePub File Size:||13.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure.
Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. Got it, continue to print. Autoclave the complete pipette without any further disassembling.
Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
To print the manual completely, please, download it. Adjustment – Easy Calibration Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions.
Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Safety Instructions Safety Instructions Please read the following carefully! Safety Instructions Please read the following carefully! Checking The Volume Checking transferppette Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
Eject bgand pipette tip.
Don’t lay the instrument horizontal when the tip is filled. Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity.
Inspect the piston and seal for contamination.
It is beyond the scope of transgerpette manual to address all of the potential safety risks associated with its use in such applications. Don’t show me this message again. Inspect the piston and O-Ring-seal for contamination. Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Table of Contents Add to my manuals Add. Pipetting Pipetting transferpegte ml and 10 ml instruments should only be used with the PE filter installed see page In order for the liquid to reach its end position, leave the tip immersed for a few seconds.
Page of 25 Go. Fitting the tip Use the correct tip according to the volume range or the color code. Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
Consult the manufacturer for use in temperatures outside of this range. To do this, affix a tip, and aspirate a sample. Enter text from picture: Don’t have an account? Page 8 Pipetting c Let the pipetting button slide back slowly.
BRAND Transferpette S Operating Manual
Also See for Transferpette S Operating manual – pages. Limitations of Use The instrument is intended for the pipetting of liquids within the following limitations: Page 17 Inspect the pipette tip cone for damage. The cycle can, however, be adjusted to individual requirements.
Pipette shaft Tip cone Fig. Hold the instrument vertically, with the sample in the tip for approximately 10 s.